to create and rate content, and to follow, bookmark, and share content with other members.
HP E4400 ---Error 513
Question asked by
on Mar 10, 2008
on Mar 14, 2008 by jborges3
Show 0 Likes
The message is 513- het VCO unlocked.
Is anyone knows how to fix this problem?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Mar 14, 2008 2:23 PM
513- het VCO unlocked
I looked this error up in the E44xx Error Guide:
Error 513 - 1 GHz Oscillator unlocked
"The 1 GHz reference oscillator is unlocked. If an external reference is
connected, check the frequency. If this error reoccurs when no external
reference is connected, the instrument may require service."
If you get this error message without an external reference connected the unit requires servicing. This error occurs when the ABUS VTUNE node goes outside of its lock range of 2V - 4V, meaning it is probably a problem with the A11 Reference Assembly.
To correct this the reference assembly will probably need to be adjusted or replaced.
Show 0 Likes
Retrieving data ...
how to simulate the large siganl loop gain
How to define explicit ports in momentum and get a valid DC solution
Using a Python Transform in WaferPro Express to Export Measured Data
Calibration Problem in NA