to create and rate content, and to follow, bookmark, and share content with other members.
3561A measurement accuracy
Question asked by
on Aug 4, 2011
Show 0 Likes
Hi all. Can someone tell me what the accuracy is for the 3561A when measuring down to -100dBm? The manual only mentioned +/-0.25dB at -51dBV. Does that accuracy apply for the entire -51dBV range?
No one else has this question
Mark as assumed answered
This content has been marked as final.
(Required, will not be published)
Retrieving data ...
Agilent Logic Analysis Service agLogicSvc.exe not responding
Cadence SiP integration into ADS
Use Momentum in Cadence Virtuoso
Uncertainty with embedding fixtures
PNA-X B model Recall print colors command not working