to create and rate content, and to follow, bookmark, and share content with other members.
8561E spectrum analyzer power menu or FFT
Question asked by
on Jun 8, 2010
on Jun 8, 2010 by mukta2008
Show 0 Likes
We have an 8561E which will not display a "power Menu" when we push the Meas/user button. It has a H02 option. the other we have has a 007 option. is this the problem? I can't find anything on the H02 option??
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jun 8, 2010 8:41 AM
Here is some information about the H02 option for the 8561E:
Absolute Amplitude Accuracy +/-0.70dB over NT&T specified freq. bands
Improved Absolute Amplitude Accuracy: +/- 0.70 dB over the
800-1520 MHz Range with the following settings:
Reference Level: 20 dBm
RF Attenuator: 30 dB
Resolution Bandwidth: 100 kHz
Video Bandwidth: 30 kHz
Log Scale: 5 dB per division
Temperature Range: 20-30 degrees C
I don't know why the power menu is missing when you pust the Meas/User button. Maybe someone else on the forum can answer it.
Show 0 Likes
Retrieving data ...
20200312_Be Ready for PCIe Gen5 Design Challenge
VEE run time error... another one.
How to define explicit ports in momentum and get a valid DC solution
Copy from[vrf]: VEEOS R1 ready for Users
PCIE Gen3 Template