to create and rate content, and to follow, bookmark, and share content with other members.
8561E spectrum analyzer power menu or FFT
Question asked by
on Jun 8, 2010
on Jun 8, 2010 by mukta2008
Show 0 Likes
We have an 8561E which will not display a "power Menu" when we push the Meas/user button. It has a H02 option. the other we have has a 007 option. is this the problem? I can't find anything on the H02 option??
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jun 8, 2010 8:41 AM
Here is some information about the H02 option for the 8561E:
Absolute Amplitude Accuracy +/-0.70dB over NT&T specified freq. bands
Improved Absolute Amplitude Accuracy: +/- 0.70 dB over the
800-1520 MHz Range with the following settings:
Reference Level: 20 dBm
RF Attenuator: 30 dB
Resolution Bandwidth: 100 kHz
Video Bandwidth: 30 kHz
Log Scale: 5 dB per division
Temperature Range: 20-30 degrees C
I don't know why the power menu is missing when you pust the Meas/User button. Maybe someone else on the forum can answer it.
Show 0 Likes
Retrieving data ...
In ADS 2015, how to define ports as internal or external while working in layout design of circuit.
ports numbers used by Agilent PC Software
Problem with allegro PCB extraction in ADS
[PDF] ADS_SI_Q&A_021 How to Import a PCB layout into ADS layout editor?
[PDF] ADS Design Flow Integration - ADFI