to create and rate content, and to follow, bookmark, and share content with other members.
how to use N7788B to measure PMD
Question asked by
on Oct 15, 2008
on Oct 16, 2008 by mkelly
Show 0 Likes
Can any one give a detail connection
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Oct 16, 2008 3:01 AM
The PMD measurements, which give DGD, IL and PDL spectra, are made with the N7788B together with a tunable laser like the 81940A or 81600B.
The optical output of the laser is connected to the laser input of the N7788B and the device under test is connected across the signal input and output connectors.
The laser is installed in a mainframe like the 8164B that is connected by GPIB to the N7788B. A trigger signal is also connected from the main frame to the N7788B with a BNC coax cable.
The measurement can be controlled from an external PC via GPIB or USB or with the internal PC of the N7788BD model.
The measurements are made during a single continuous wavelength sweep of the tunable laser with an enhanced implementation of the Jones Matrix Eigenanalysis method.
More details are also in the User's Guide, located here:
Show 0 Likes
Retrieving data ...
Unleveled source error message
Agilent 33500A/33500B function gen Binary ARB example program
Save a screen capture and a waveform file remotely with a scope.
[2019Q2]Automotive Radar Workshop
汽车雷达系统设计与仿真高级进阶系列-2: 非线性FMCW信号生成及其影响 Non-linear FMCW Signal Generation and Its Impact on Automotive Radar System Performance