to create and rate content, and to follow, bookmark, and share content with other members.
Can the X-Series measure Bluetooth hopping signals?
Question asked by
on Nov 28, 2011
on Nov 28, 2011 by tabbott
Show 0 Likes
Can the X-Series signal analyzers with the Bluetooth measurement app (N9081A) measure the Bluetooth hopping signals?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Nov 28, 2011 12:24 PM
Hi Gladys -
The short answer for this question is “no”.
Fundamentally, frequency hopping is needed for testing the functional capability of the Bluetooth device, whereas for parametric tests, hopping is not essential.
To reduce the number of variables and identify individual performance characteristics, frequency hopping is turned OFF for a number of tests.
The N9081A and W9081A Bluetooth measurement application supports the 79 MHz transmit bandwidth transmitter tests, but does NOT support hopping signals.
Show 0 Likes
Retrieving data ...
how to simulate the large siganl loop gain
How to define explicit ports in momentum and get a valid DC solution
Using a Python Transform in WaferPro Express to Export Measured Data
Calibration Problem in NA