Ok, I am calibrating an 8593E spectrum analyzer, with option 140. It is failing the Marker Count portion of its performance tests at 1.5GHz only with 20MHz and 1MHz spans. I have already verified the accuracy of the 10MHz reference and the 300MHz cal output. I have also already performed the Sampler Match adjustment. Any ideas how to fix this problem?