to create and rate content, and to follow, bookmark, and share content with other members.
Characterize 80+80 MHz signal with single-channel analyzer
Question asked by
on Jul 11, 2011
on Jul 11, 2011 by tabbott
Show 0 Likes
How can I characterize an 80+80 MHz signal, if I only have a single-channel analyzer?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jul 11, 2011 4:08 PM
Each frequency segment of an 80+80 MHz signal is completely self-contained. Each frequency segment just contains a standard 80 MHz 802.11ac signal, and the 802.11ac transmitter requirements like constellation error and spectral flatness apply separately to each segment. Therefore, a single-channel 80 MHz analyzer can be used to analyze and characterize each segment separately.
Of course, analyzing the segments separately does not provide any information about time alignment, frequency drift, or interactions between the frequency segments. A multi-channel analyzer is needed for this.
Show 0 Likes
Retrieving data ...
3325B Fail 26 Error - how to fix
shematics for 8590L A16 board
License Installation and Environment Setup
How to use Cadence Schematic Design in ADS - DynamicLink
ADS2009 Legacy User Templates