to create and rate content, and to follow, bookmark, and share content with other members.
Question asked by
on Apr 18, 2011
on Apr 18, 2011 by tabbott
Show 0 Likes
How to save the "Corrected Flatness" data for the frequency response adjustment in the ESA-L1500A if not using the adjustment software?
Does anybody know the prrocedure for that?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Apr 18, 2011 1:48 PM
The ESA-L1500A could only be adjusted via Agilent software. The frequency response adjustment test came up with the corrections factors and then stored those corrections to analyzer memory. I haven’t heard of a manual procedure. In fact, a big majority of test equipment today cannot be adjusted manually; it requires the appropriate software.
Show 0 Likes
Retrieving data ...
circuit component pin number does not match EM layout pin number
How do I have specific BJT model
Is there a Step recovery diode model for ADS2016? There is a design kit for download ADS2005A, but when installed in 2016, it says design kit does not contain lib.defs file.
port solver warning
How to define explicit ports in momentum and get a valid DC solution