to create and rate content, and to follow, bookmark, and share content with other members.
89600 VSA SW crashing when a turning timebase **** on scope
Question asked by
on Oct 9, 2008
on Oct 9, 2008 by mlombardi
Show 0 Likes
Have you heard anything about 89600 VSA SW crashing in 90000-Series scopes when the user turns the timebase ****?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Oct 9, 2008 3:13 PM
The general warning in the Infiniium/VSA product note is that you should not adjust any front panel scope controls while the 89601A VSA SW is running. All oscilloscope functionality should be controlled through the VSA SW.
The procedure to control the oscilloscope is listed in the help system and included below.
Front Panel Scope Control: You can use the Disconnect feature to pause the 89601 and release its control of the Infiniium Series scope. Then you can use the scope independent from 89601. When you go back to the 89601 and start or resume a measurement, the 89601 restores the scope state that was set before you disconnected.
If you are doing cross domain analysis and want to measure the same information in both the time domain and the modulation domain the recommended procedure is to set VSA to single, when the signal is captured the VSA will pause. Disconnect VSA and then "alt-tab" to the oscilloscope display, the capture in VSA is the same as the one captured by the oscilloscope.
Show 0 Likes
Retrieving data ...
Changing Start-up directory of the ADS 2015.1
Writing and debugging AEL Expressions
List all equations in ADS data display window and locate the selected one 在ADS数据显示窗口列出所有公式，并定位选中的公式位置
54645D Power Supply Board (0950-2735)