to create and rate content, and to follow, bookmark, and share content with other members.
Function of a airline
Question asked by
on Apr 15, 2011
on Apr 15, 2011 by Dr_joel
Show 0 Likes
Can someone help me and explain what the heck is the function of an airline. Is it a precision termination, vswr, precise distance, or what.
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 2 comments
(Required, will not be published)
Apr 15, 2011 9:41 AM
There are several functions of an airline (listed in reverse order of usefulness):
1. They help you fly from point A to point B! (but I don't think this is what you are talking about)
2. They are sometimes used just as a low-loss physical extension
3. They are used to add line length to a measurement (helps to add ripple on reflection measurements)
4. They are a
impedance reference for the length of the airline.
Item 4 is the most important and is the whole purpose of the airline...unless you need to travel! They can be made with great precision to be exactly 50 ohms impedance (typically). They also have very low loss. These characteristics are used to help measure/determine specifications at higher frequencies where a fixed load is inadequate. Because of the great precision and stability of well-made airlines, they can be used as 50 ohm transfer standards.
While airlines typically have a precise electrical length specified, this is not a major concern for most applications.
One common application (#3 above) is to use an airline to determine source match. A reflection measurement is made with an airline attached. Some known mismatch is connected to the end of the airline. This will produce ripple in the resulting measurement and, based upon the peak-peak amplitude of the ripple, the source match can be determined.
Show 0 Likes
Apr 15, 2011 6:42 PM
As Jvall said, the airline is a precision impedance transmission line. As such, it's S11 is zero, and S21 is nearly 1. So, it provides phase change with no amplitude change. If you do a 2 port cal, then re-measure the thru, it's always flat (unless you use unknown thru, then it's the thru loss). If you measure an airline, it should also be perfectly flat, but it will always show ripple. The ripple is a measure of the badness of the calibration.
Show 0 Likes
Retrieving data ...
GPIB/SCPI Commands wtih Agilent E8361A
SystemVue Connected Solution (Connected to Keysight Instruments)
Connecting SystemVue to Test Instruments for DPD Measurement
How to choose the right optical power meter for 850 nm signals
On a E4438C when using a PRAM file for data I was able to add a data bit with non burst too turn off the carrier. I can not seem to do this with the N5182B using the same PRAM file. How do I turn off the RF?