to create and rate content, and to follow, bookmark, and share content with other members.
Tracking Generator on N9340B
Question asked by
on Nov 6, 2009
on Jun 4, 2010 by mukta2008
Show 0 Likes
Is it possible to use the tracking generator as a simple sig gen for testing receiver sensitivity, it would save me carrying another instrument.
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jun 4, 2010 2:44 PM
You can use the tracking generator of the N9340B for the purposes of outputting a CW signal if that is what you intend to work with - a CW signal output for your receiver test. The way to accomplish this is by setting the span to 0Hz in the Tracking Generator mode and setting the center frequency at the desired CW frequency, then setting the amplitude at a desired level, and finally turning on the amplitude.
Show 0 Likes
Retrieving data ...
Inductance value from S-parameters
IVI Driver for N7714A Four-Port Tunable Laser System Source
20200312_Be Ready for PCIe Gen5 Design Challenge
SystemVue support import CST antenna pattern file only suffix for FFS file format?It can support other format file?
Modeling of GaN with $table_model in ADS using verilog-ams