to create and rate content, and to follow, bookmark, and share content with other members.
Can N9912A FieldFox be used for viewing pulsed GSM signals?
Question asked by
on Nov 24, 2008
on Nov 25, 2008 by am95405
Show 0 Likes
Can you please explain how the N9912A can be used in viewing pulsed GSM signals?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Nov 25, 2008 9:26 AM
FieldFox N9912A's spectrum analyzer mode can be used to measure GSM signals. You may need to change the sweep acquisition mode from Auto to Manual. This setting can be found under the Sweep (3) key. When set to Auto, Sweep Acquisition is set to the minimum value required with the current settings in order to achieve amplitude accuracy when measuring CW signals yielding the fastest sweep update rate. However, you can adjust this setting in order to increase the probability of intercepting and viewing pulsed RF signals. While watching the trace, increase the SwpAcquisition value until the pulse spectrum rises out of the noise and reaches its maximum level.
Show 0 Likes
Retrieving data ...
Earth Day 2018 and Natural Resource Conservation Learning. Not Just for the Next Generation.
What is Bandwidth? How Much Do You Need?
mapping "purpose" for a layer
Register for the Keysight Community and Follow a Forum or Blog
8563ec error 319