to create and rate content, and to follow, bookmark, and share content with other members.
35665A Dynamic Signal Analyser - Bus Error
Question asked by
on Dec 16, 2008
on Jun 1, 2010 by mukta2008
Show 0 Likes
I have a 35665A signal analyser that fails at boot up with the message BUS ERROR FC=5 RW=0 SZ=1. Any ideas how to troubleshoot this?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jun 1, 2010 8:06 AM
Typically, the information to troubleshoot these types of errors is in the Service Guide for the analyzer. If you have the 35665A service guide, there is a separate troubleshooting section in it which will address this type of an issue. Agilent doesn't have an online version of this manual but you can buy a hard copy of it from companies such as Manuals Plus (
) who have a repository of old HP/Agilent manuals.
To give you an idea about the contents of the service guide, please refer to 35670A (successor to the 35665A) service guide available online at
http://cp.literature.agilent.com/litweb ... -90066.pdf
Show 0 Likes
Retrieving data ...
Uncertainty with embedding fixtures
How to trigger FieldFox when average on with SCPI commands
1 path 2 port cal vs. Enhanced Response cal
Measuring Thermal Noise of Amplifier
HP 8594A YTO readings