to create and rate content, and to follow, bookmark, and share content with other members.
8163B Optical Powermeter
Question asked by
on Feb 13, 2007
on Feb 13, 2007 by mkelly
Show 0 Likes
What is the maximum number of data points for 81635A in the "logging" mode?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Feb 13, 2007 1:44 AM
When the instrument is programmed remotely via GPIB, the maximum number of measured data points is 20,001 for most of the 816x power meters. The Fast Power Meters, 81636B and older 81637B can take 100,001 data points.
If the power meters are providing only 10001 points, this can be improved by upgrading the module firmware to the latest level, as available at
When making spectral measurements with the tunable laser, the laser will deliberately overshoot the start and stop wavelengths by at least 50pm or more. This can result in the power meter needing space for more points than just those within the chosen wavelength region. For example, sweeping over 1520nm-1570nm (50nm) with 1pm resolution requires 50,001 data points plus 100 data points for overshoot.
Show 0 Likes
Retrieving data ...
New vs. Old IO Libraries Questions.
SCPI 34970A Connection Exception
AEL in ADS 2012.08 (New User to AEL)
# Query for E4433B ESG-D Series signal generator regarding 2GFSK.OQPSK Signal generation
Keysight IO Libraries 2018 Suite cause hanging on Windows 10?