to create and rate content, and to follow, bookmark, and share content with other members.
How do I get an 82357B to work?
Question asked by
on Jun 6, 2007
on Jun 6, 2007 by Brig Asay
Show 0 Likes
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jun 6, 2007 10:47 PM
The new 82357B USB to GPIB adapter is out. It's very nice and USB 2.0 but you need a secret driver to get it to work with the Agilent I/O libraries 14.2. This is the latest version of the I/O libraries and makes no mention of the extra driver you need. The 82357B web site also makes no mention of this driver, nor does the 82357B manual. It is available at the following link:
So to get an 82357B (the 'A's are now obsolete) to work you need to install the I/O libraries version 14.2 and then the 82357B driver. Do this before you plug in the 82357B and it works great.
Show 0 Likes
Retrieving data ...
Parameter-sweeping the endpoints of a waveguide port
How to find non-interpolated S-parameters in EMPro?
Scope firmware for 90000 Series;Windows XP operating system It is running 1.20 now. upgrade to the latest software is 4.20
Get Trigger Happy with Keysight Infiniium Oscilloscopes