to create and rate content, and to follow, bookmark, and share content with other members.
Multiple readings of waveform
Question asked by
on Oct 25, 2010
on Nov 2, 2010 by algoss
Show 0 Likes
How do I take 10000 readings of a waveform and then obtain lowest frequency, highest frequency, lowest period, and highest period using Agilent DSO 80204B?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Nov 2, 2010 6:02 AM
First, it would help, if we knew more about your waveform.
Beyond that, there are a few solutions. If you need exactly 10000 measurements, then you will have to remotely control the scope. If the number of measurements just has to be larger than 10000, then turn on a Period and Frequency measurement, and let the scope run until you have accumulated more than 10K measurements. The stats at the bottom of the screen will tell you the largest and smallest of both measurements. If you have EZJIT, then you can capture that data much more quickly, because it will make measurements on every cycle of every capture. If the signal has a high enough frequency, and you have enough memory depth, then it might be able to do it in a single capture.
Show 0 Likes
Retrieving data ...
Budget functions in schematic
circuit component pin number does not match EM layout pin number
Simulation error:error detected by hpeesofsim in loading dynamic device code during netlist flattening.
ADS netlist – error caused by duplicate expressions in export