to create and rate content, and to follow, bookmark, and share content with other members.
Question asked by
on Sep 27, 2007
on Oct 8, 2007 by marowley
Show 0 Likes
What is the rate or aquisition time for oscilloscope DSO-081204B to take waveform for averaging operation?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Oct 8, 2007 1:50 PM
This question is truly very vague.
I assume â€œrateâ€ is defined as Waveform Update Rate
To answer this accurately there needs to be a better definition about the signal that needs to be averaged. There are many variables that will determine how long it will take the oscilloscope to finish averaging.
To name a few:
Memory depth needed?
Sample rate needed?
Input signal repetition rate or frequency?
How many averages are selected?
Horizotal timebase setting?
Waveform update rate
Fastest 4,800 waveforms/sec (memory depth: 64 pts, sampling: 10 GS/s, time/div: 50 ps,
connect dots: off, Sin (x)/x: off, color grade: off)
Nominal 800 waveforms/sec (memory depth: 1000 pts, sampling: 40 GS/s, time/div: 500 ps,
connect dots: on, Sin (x)/x: on, color grade: off)
Show 0 Likes
Retrieving data ...
Outdoor RF Sensors (N6841A)
Unable to install Command Expert Command sets
A list of questions about the N6841A RF sensor
HP3458A err message