to create and rate content, and to follow, bookmark, and share content with other members.
Question asked by
on Sep 27, 2007
on Oct 8, 2007 by marowley
Show 0 Likes
What is the rate or aquisition time for oscilloscope DSO-081204B to take waveform for averaging operation?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Oct 8, 2007 1:50 PM
This question is truly very vague.
I assume â€œrateâ€ is defined as Waveform Update Rate
To answer this accurately there needs to be a better definition about the signal that needs to be averaged. There are many variables that will determine how long it will take the oscilloscope to finish averaging.
To name a few:
Memory depth needed?
Sample rate needed?
Input signal repetition rate or frequency?
How many averages are selected?
Horizotal timebase setting?
Waveform update rate
Fastest 4,800 waveforms/sec (memory depth: 64 pts, sampling: 10 GS/s, time/div: 50 ps,
connect dots: off, Sin (x)/x: off, color grade: off)
Nominal 800 waveforms/sec (memory depth: 1000 pts, sampling: 40 GS/s, time/div: 500 ps,
connect dots: on, Sin (x)/x: on, color grade: off)
Show 0 Likes
Retrieving data ...
How to Import W-Element Table Model into ADS
Optimal FEM padding for broadband FEM simulations (EMPro, ADS)
ADS momentum layout optimization
Inductor Design Workshop using ADS
SystemVue 2017 Leads the 5G Design and Verification Process