to create and rate content, and to follow, bookmark, and share content with other members.
Cal Set And VI
Question asked by
on Oct 11, 2010
on Oct 12, 2010 by bhokkan
Show 0 Likes
How can I apply a Cal Set to the channel, with the SCPI commands, if I haven't got the SENS:CORR:CSET:ACT command.?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Oct 12, 2010 8:02 AM
Hello. If your PNA does not have the
command then I believe it must have Firmware Revision A.05.25 or lower, because that command was introduced in the A.06.04.xx firmware versions. The command that was used in those previous revisions for selecting a calset is
. Note that the topic (in the above link) for the ACTivate command mentions that it 'supersedes' that command. GUIDs were originally intended to be the way to ensure each calset would always have a unique identifier, but in the more recent years we just began ensuring the calset names had to be unique also, and provided name-based commands to make remote use/manipulation of calsets a little easier.
is the command you can use to get the list of GUIDs of the calsets you have; note that the optional argument ('NAME' or 'GUID') was added in the newer firmware revisions, so with your revision you do not pass an argument to the query and it always returns the calsets GUIDs.
The other way to select a calset into a channel is the 'heavy-handed' way which is to load your saved calibrated instrument state file (
). Note that with Firmware A.05.25 and lower, you would use CST ('.cst' files) for that, because the '.csa' file format ('CSAR' command) was introduced in firmware revisions A.06.04.xx and higher.
Show 0 Likes
Retrieving data ...
33522B - CFG011
How disable secure environment option (SEC) on MSO7000A?
ADS Python Data Link 安装设置指南
ADS Crash now the workspace/library is stuck "read only"
Hello, I am getting these two warnings, I was wondering what do they mean and how serious they are: 1- Matrix is singular (detected at node or branch 'L2.i'), circuit converged after a virtual resistance of 1 Tohms bet each node and ground, a virtual resi