how to set up occupied bandwidth on n9020
The answer to this problem, as well as much more information pertaining to it, lies within the Help files of your MXA. This can be found at https://www.keysight.com/main/editorial.jspx?cc=US&lc=eng&ckey=2431261&nid=-32508.1150426.00&id=2431261 . I found this solution in the Occupied BW section of the .chm file at this link.
As for a simple answer, here are the steps:
1. Select the tab at the top of the display, and choose the Spectrum Analyzer mode.
2. Select the Occupied BW measurement under the Measurement section
3. Choose your view (Results, if you would like to quickly make the measurement)
4. Go back to the main display and record the measured values
Retrieving data ...