For my measurements with E4990A, I use 3 channels. Does I need to compensate (open and short) for each channel - or compensating for 1 channel is enough?
I'm looking for internal command that shows me when my measurement, in specific channel, has ended.
I need it for a macro, to trigger the next channel measurement.
Where I can get more information or examples for internal commands?