In 66319D when we turn on the output getting unregulated error.what could be the reason for getting this error.
I already had this error on my 66319B, it was just a submin fuse on main board (FB761, Keysight part number 2110-0712, equivalent Digikey part F5566CT-ND )... maybe it's the same problem.
I'm getting this same issue one of my N6952A supplies hooked to a Keysight electronic load (N3300). When the output is first enabled, I get UNR from the Ques Status Event Register for a second or two, then it goes away. A second N6952A hooked to a different N3300 channel works as expected.
Retrieving data ...