how do I have to process the real attenuation of my 11708A reference attenuator with 29.86dB (nominal 30dB), when I perform the ZERO/CAL routine of the power meter?
I can use the CHANNEL Offset and set it to a value of .14dB. After performing the ZERO/CAL procedure, do I have to clear this generally offset for the measurements?
Do I have to clear the offset for measurements outside of 50MHz?
Thanks and regards