Hi at all,
does anyone have detailed information about the Selftest of the L4534A Digitizer? How does the self test works?
The overvoltage protection test is sometmes failed, if the unit is in warm conditions (powerd for several hours),
in cold case, the test is okay.
Where is the internal calibration source laoceted? Is there one for each channel board or only one on the main
Thanks and regards