to create and rate content, and to follow, bookmark, and share content with other members.
Balun measurement with 85071B
Question asked by
on Apr 11, 2007
on Apr 11, 2007 by Dr_joel
Show 0 Likes
How to measure balun parameters with a 4 port 85071b
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Apr 11, 2007 9:19 AM
There is a fixture simulator that is in the ENA, which makes balun measurements very easy. In the fixture simulator, choose the SE-BAL condition.
Before, make a 3 port cal on the ports to which you want to connect your balun.
Your balun loss is Sds21, your balun match is Sdd11. You can also show the results with other impedances, if your balun is not 100 ohms on the differential side and 50 on the Single-Ended (SE) side by using the Port Impedance settings. Set the impedance on the two ports that are used as the balanced port to each be 1/2 of your desired balanced impedance.
Show 0 Likes
Retrieving data ...
Earth Day 2018 and Natural Resource Conservation Learning. Not Just for the Next Generation.
What is Bandwidth? How Much Do You Need?
mapping "purpose" for a layer
Register for the Keysight Community and Follow a Forum or Blog
8563ec error 319