I am working on E5071B, but when port calibration using female calibration kit the result can not flat 0.0 for the log mag ( screen captured). Is this normal ? Or need to repair for the calibration kit ?
Additional info, when I am using malem calibration kit the result can 0.0 flat for the log mag. Please help me. Thankyou very much.