Digitizer Adapter for E5503B Phase noise system. We would like to buy the Digitizer Adapter for E5503B phase noise system. Is the part number E5500-60003? Do you still sell them? How much is that?
It does seem that E5500-60003 is what you're looking for. Note that you can search for parts with our Find-A-Part tool (works best on Internet Explorer). We don't sell this adapter the normal way anymore, but I recommend that you call our support specialists at the number found in the "Contact an Expert" link on the product page for the E5503B to see if there's anything they can do to get you this part and what that might cost.
Thanks for the info
We don't use the old digitizer board for this system. You could use a 89410a or E4401A or E4402A as an external analyzer which actually works better than the digitizer.
Can you elaborate how to use a 89410a or E4401a or E4402a as an external analyzer instead of Old Digitizer Board?
Thank you in advance.
The P/N software allows you the choice of which analyzer you want to use for your measurements. The old digitizer board would only do offsets .001 Hz to 100 KHz. The 89410A will do .01 Hz - 10 MHz. The E4401 or 2 with option 1DR and 1D5 will do offsets of 1 Hz - 100 MHz. 100 MHz is the limit of the PN heads.
Thank you very much for your reply. Do you know on how to setup E4401 or 2 for E5503B system?
Thank you very much.
You only need to select the model you are using. Select the address the instrument is at. It should not conflict with any other instrument. Connect the analyzer IF output out of the 70420A to the RF input of the Spectrum analyzer. Connect the time base from the 70420 system to the spectrum analyzer time base in. That's all there is to it.
if we use spectrum analyzer, does it mean it will bypass the Phase Noise Test set?
No, the spectrum analyzer is simply does the IF measurement as an alternative to the digitizer board. Based on your offset measurement requirements. The external spectrum analyzer is required for any offset measurements >100KHz. We use our 89410A as we have testing from 1 Hz to 10 MHz offsets we need to measure. Keysight has a downloadable User's Guide that shows how to connect up the instruments for proper measurement. Exactly what are you using this system to measure?
OK. Thank you. I am try to measure phase noise for our Synthesizer that has output Frequency 15 GHz. I would like get plot for offset measurements > 100KHz. Can you share the link of the Keysight's downloadable user's Guide that show how to connect up the instruments for proper measurement please?
If you go to the product page on the main website. Then on the product page you will see a tab for manuals and documentation. Under that Tab a list of downloadable documents will come up. You will find what you a looking for there.
Retrieving data ...