Just purchased 16454A Magnetic Material Test Fixture.
Is there a way to use 1645A test fixture with PNA Network Analyzer N5225A ?
If Yes, how can I set that up ?
If Not, what analyzers should I use ?
I took a look at the brochure for the 16454A Magnetic Material Test Fixture and found this table which may help you:
It looks like your test fixture is designed to work well with the E4991B and the E4990A.
If you need more information, don't hesitate to contact a product selection expert by clicking on "Contact an Expert" on the product page for the 16454A and using one of the contact methods shown on the Product Selection Experts tab to ask a question.
Thanks for the information, was looking for that
Whilst not supported on a PNA, there are almost certainly ways you can get around it.
Whilst not stated in that document, I'm pretty sure it is supported on the 1.8 GHz HP 4291B I own - well technically, my company,. Kirkby Microwave, owns it. If I did not have such a hangover, I would look! You basically calibrate at an APC7 plane, then add the fixture. The 4291B works on the RF-IV principle, rather than the bridge of a VNA.
The radio amateur Kurt Poulsen email@example.com (OZ7OU) has done a lot with making fixtures similar to the HP/Agilent/Keysight one, and using them with the low cost VNWA, which is a 1300 MHz VNA. I'm sure if you contact Kurt, you will find how to use such a fixture with a vector network analyzer. He has written some material on it, but again an overdose of C2H5-OH last night means I don't feel like hunting for the material. Kurt is very helpful.
I just sent a note to Kurt. If he has access to these forums, which he may do, perhaps he can post a link for others to see.
I think I just need to go to bed now!
Thanks drkirkby, I wasn't aware of that.
Retrieving data ...