to create and rate content, and to follow, bookmark, and share content with other members.
8753 down to 30kHz
Question asked by
on Dec 6, 2010
on Dec 7, 2010 by Dr_joel
Show 0 Likes
Is it just the 8753's with built in test sets that go down to 30kHz? My B,C and D with external test sets only go down to 300kHz. I just started a new job today and will be dealing with LF,MF and HF. I'm used to >=VHF.
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Dec 7, 2010 7:55 AM
yup, different bias tee and particularly different bypassing (that's the tricky part). Performance down to 30 kHz is a bid dodgy, the match goes quite bad but a such low frequencies, often match is not so much of a concert. In fact, the operation supports 10 kHz but the performance is really quite bad down there (like 4 db return loss).
Show 0 Likes
Retrieving data ...
Thinking how to get money off Cash App without card and without net? Call customer care.
how to simulate the large siganl loop gain
Contact Facebook if you have problem getting a code for account activation?
Power calibration with use of receiver "A"
Importing EMPro circuit component definition from ADS