to create and rate content, and to follow, bookmark, and share content with other members.
Question asked by
on Oct 18, 2006
on Oct 18, 2006 by Dr_joel
Show 1 Like
How to measure the attenuation coax cables (attenuation=30dB/100m @ 1GHz) with 8753E analyzer? Noises are begun from 300MHz!
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Oct 18, 2006 8:30 AM
With long cables, or other electrically long devices, you must be careful of a problem we call "I.F. delay". Here's a good note on this:
Simplest way to fix this: use 30 Hz IFBW on 8753.
use 1000 Hz IFBW on PNA
If you don't want to use 30 Hz, shoot me an email and I'll tell you other alternatives
1 person found this helpful
Show 0 Likes
Retrieving data ...
[WMV] 介紹CMOS OP放大器的DC, AC及Transient模擬
Two Types of Antennas You Can Design With ADS
How to draw structure in in Momentum and EMpro by paramters
Antenna Design Using Fully Integrated 3D EM in ADS
Adding a Port to an Antenna and Moving one antenna relative to another