to create and rate content, and to follow, bookmark, and share content with other members.
Measure waveguide device on N5250A 110G NA with coax port
Question asked by
on Oct 15, 2007
Show 0 Likes
Apart from using coaxial-to-waveguide adapters, is it possible to access the WR10 port of the mm-wave head before it emerges from the combiner into a coaxial test port?
No one else has this question
Mark as assumed answered
This content has been marked as final.
(Required, will not be published)
Retrieving data ...
Uncertainty with embedding fixtures
PNA-X B model Recall print colors command not working
Ports in Momentum: SMD vs TML
A Simple Method to Verify the Bandwidth of your Probe