We are using Keysight 34465A 6 ½ Digit Multimeter for testing power consumtion of low power sensor. But we got stuck on calibration of device. I saw this e-mail in keysight forum and wanted to consult you about this topic.
Our device’s approximate current consumptions;
- In sleep mode, 5- 200uA
- In normal mode, 3-5mA
- In transmit mode, for low power: 100-120mA, for high power 900-950mA
- In receive mode, 10-15mA
So I want to analyze power consumption of this device. I used digitizing method with 1ADC dynamic range. In results, i got so many minus values up to -500 uA. So, how can i compansate these values. I will send 1 screenshoot as a attachment. Can you help us about calibration ?