What remote commands are needed to use pin 1 and 2 of the digital I/0 as outputs.
And what is the specific command to trigger the output to 1(TRUE) for some ms?
Command to send to use 3 pin for output with positive polarity.
:DIG:PIN1:FUNC DIO 'in this example, the configuration is for the pin 1 only
:DIG:PIN1:POL POS 'positive polarity for pin1
:DIG:OUT:DATA 3 'output pin 1 and 2 are set to 1 (true), if pin was set to DIO function previously.
:DIG:PIN2:FUNC TOUT 'in this example, the configuration is for the pin 2 only
:DIG:PIN2:POL POS 'positive polarity for pin2
: LIST:TOUT:BOST 1, (@1) 'set list of 1 point, and a trigger will be send at pin#2, when output channel#1 will be set to ON.
I don't have an E36312A, I found these informations on programming manual. For TRIG command, I'm not sure if everything are OK.
I hope these informations may help you.
Retrieving data ...