How to use the PNA of N5247B and the optical measurement system to measure
the delay of a fiber with 5 meters ?
so, I presume you have optical transducers? If so, you can just connect a very short cable between the transducers, normalize the trace, then put on your fiber under test and change the S21 format to delay.
You do have the optical tranducers (optical modulator or laser, and optical receiver or photodiode)?
Retrieving data ...