Here is my scenario:
* I am collecting data with "BenchLink Data Logger 3" from two units of "34970A".
* I tried to collect data over 24 hours ( data size will be smaller than 10 MB), and want to monitor the data anytime.
* Unfortunately, BenchLink Logger will lost connection to both 34970As (at different time) after a couple hours.
(although 34970As are still running by themselves)
* I know 34970A can run by itself without connection to PC.
* But when I tried to upload the data from 34970A, it reports that channel configurations are different from latest channel configurations on BenchLink Logger, how come?
Anybody has similar problems? any hint?