Hello. I am using PNA-X N5242A, the multichannel VNA. Now I don't know how to configure Labview to read out the multichannel signal like S21 and S41 at the same time? Are there any examples about it? #VNA#N5242A
From your question, I take it this might be your first venture into measurements and data acquisition with a PNA-X. Unfortunately, we don't have any published Labview examples, but if you can send SCPI commands from Labview to your instrument, then you can use all the examples we do have in our online help. Specifically, take a look at examples in the "Setup Measurement" section and ones under "Miscellaneous" titles "Getting and Putting Data".
You setup a single channel to have two traces. Depending upon what you are measuring, you can choose different calibrations. Then you will have a trace of S21 and a trace of S41. You can save the data to CSV and then open the data file (super easy) or send SCPI commands to read back the data into lab view (not terribly hard, the scpi commands show you in the examples). What's the application?
Retrieving data ...