E4990a - Hi all, how to change Sweep direction parameters automatically, up to down and immediately down to up, in two following single measurements ? thanks
Basically, E4990A does not have the direct function/feature to do so although you can change the sweep direction on E4990A using front panel key.In order to change sweep direction automatically in two single measurements, you can create a program that utilize the following SCPI command. You may refer to the link below for more details on the command.
Thanks & Regards,
Retrieving data ...