to create and rate content, and to follow, bookmark, and share content with other members.
4395A low freq. Z Measurement
Question asked by
on Jan 4, 2010
on Jan 5, 2010 by RayH
Show 0 Likes
I want to measure impedance down to 9kHz. I have a bridge that works to 100kHz, so not wanting to have to purchase an 87512A. Can I use an older 35676B which would appear to do the same job, but to max 200MHz and 75ohm?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jan 5, 2010 8:45 AM
The 35676A/B work on the same principle as the 87512A/B. It should work OK with the 4395.
I don't think the spacing of the connectors on the 35676 will fit the front panel of the 4395A, but you can use cables to attach the bridge. One point to remember is you must perform a "one-port" calibration (using the open, short, and load standards) to get meaningful data when using this bridge.
Show 0 Likes
Retrieving data ...
# Query for E4433B ESG-D Series signal generator regarding 2GFSK.OQPSK Signal generation
34972a; Read latest data of multiple channels during continuous scan while writing to USB drive
Anybody knows how to simulate a Verilog-a file defined FET in ADS?
Importing Orcad Schematic to ADS
ADS Netlist – export, edit, import/simulate