In the 34461A DMM in the table "performance versus measurement speed", we can read for a 0.2 PLC 100 Readings/s, but where does it come from? Because 3 ms of integration time + 1ms of delay (immediate sample) give us 4 ms so that 250 Samples/s.
My question is :
What the time between two samples when we have 0.2 PLC with 1ms in the trigger delay ?
Is the sampling time constant during the acquisition if the calibre is in manual but with the autozero ?
Thank you for your help