I'm measuring mixers and have an issue displaying the input frequency range for S11 measurements. The display will always show the output frequency range. Is there a fix for this?
Select the S11 trace and change the x-axis
if you have multiple traces, only the ones you select and change the x-axis for will show up with the input annotation:
this all depends on the fact that you must have a swept input range. Any swept range can be used as the X-Axis annotation.
Ahh very easy, thanks for the help!
Retrieving data ...