I have a 8722ES with the direct sampler access option.
I would like to use a GaAs MMIC step attenuator in the sampler path. However, I have noticed that there is 0.006 V of DC offset at the step attenuator output (when measured unloaded with a multimeter). I am wondering if this present a danger to the sampler input or if I can connect this attenuator directly to the sampler input without risk.
Thank you for your help,