i use Daq Agilent 34970A, i have a problem, every time i push the scan button (to save) then after 1 seconds (after one sound bib), the scan is automatically turn to off. i try many times, and the results is same.
Please help me on this issue..
I assume that you have configured multiple measurement channels. The default trigger source is immediate and the default trigger count is 1. If you want continuous scanning after you press the Scan button, you must set the trigger count to Continuous using the Interval button. With the Interval button, you can set the time interval between scans and the number of scans or continuous.
Retrieving data ...