33220A arbitrary waveform generator. I'm getting a negative pulse at power up, 2 seconds long, about -4V, prior to the output going high impedance. It is set to power up to the default state. Is there a way to prevent that negative pulse?
Hi, I tested that on my 33220A and I saw nothing like that...start with default configuration. Firmware revision 2.07-2.02-22-2. Maybe you have a problem with your unit...I don't know !
Retrieving data ...