Can the line volted be changed from 120VAC to 240VAC?
Are there ant internal taps or links inside the unit so you can do this?
You can change from 120VAC to 240VAC. There are S400 and S401 Voltage selector inside, on main board of 6643A (inside, near front panel at left).
Hi there and many thanks for your reply.
Yes I have sorted it out but there was not a switch panel so I had to remove the transformer and sort out all the connections that where not being used.
It all works fine now.
Thanks again for your help.
Retrieving data ...