to create and rate content, and to follow, bookmark, and share content with other members.
8510 50GHz w two sources ?
Question asked by
on Jan 7, 2009
on Jan 8, 2009 by dgun
Show 0 Likes
i have a 8510C with a 83631A and a 8515A testset, now i got offered a 8517A testset which would get me to 50GHz, i also have a 83555A source module that covers 33-50GHz can i marry those together to get 50GHz coverage ?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jan 8, 2009 11:23 AM
Theoretically, you can do it if the source has the module interface (I think all 8360s had this). But you might have issues with the 8510 knowing what the available frequencies are or it might set the 83631A so that the 83555 doesn't work properly.
Of course, when the 83555 is connected, you would lose the low end of the frequency range, and the 83555 has much worse ALC range (only 8dB) than the 83631A (24-30 dB).
Show 0 Likes
Retrieving data ...
Oscilloscope not connected in Connection Expert
Agilent Logic Analysis Service agLogicSvc.exe not responding
Cadence SiP integration into ADS
Use Momentum in Cadence Virtuoso
Uncertainty with embedding fixtures