We are using the AFR to calibrate out the test fixture we are using.
After completing the calibration and applying corrections, we noticed that our return loss is above 0. (No DUT connected)
Attached the CSA file with no AFR and AFR applied. PNA-X firmware version is A10.40.14.
Did I missed out anything during calibration?