to create and rate content, and to follow, bookmark, and share content with other members.
4 Port PNA-X test Mixer IMD
Question asked by
on Feb 27, 2016
on Feb 29, 2016 by daras
Show 0 Likes
When 4 port PNA-X test Mixer IMD, Need an external source? Can give me the test diagram? Thanks, Yezhou
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Feb 29, 2016 9:02 AM
If you are testing a mixer with a single LO and you want to measure IMD, then you do need an additional external source. the 4-port PNA-X can be configured to use the external source either as the LO for the mixer or as the 2nd tone (F2) for the IMD measurement. the most simple configuration is to use the external source as the LO. if you don't have to measure the LO port match, then you simply connect the mixer between ports 1 and 2 of the PNA-X and you connect the external source's RF output directly to the mixer LO port. you will still need to connect the 10 MHz references of the PNA-X and source together and you will need setup the external source as an external device on the PNA-X, so that the PNA-X can control the source and set its power and frequency(ies). If you do need to also measure LO match or any kind of LO to RF or LO to IF isolation, then that setup is also possible, but it is a bit more complicated. let us know if you do have these additional measurement requirements, and we'll be happy to provide a simple block diagram of the measurement setup.
Show 0 Likes
Retrieving data ...
How to solve problems in E3631A Power Supply?
Does Benchvue work with E4440A, and HOW?
[PDF] Signal Integrity Q&A Collection using Keysight ADS
IO Library stopped after Windows update. What´s wrong ?
IO Library 18 did not install properly