to create and rate content, and to follow, bookmark, and share content with other members.
34461A DMM and temp probe use.
Question asked by
on Jan 27, 2016
on Jan 27, 2016 by LA7SJA
Show 0 Likes
Just want to know if there is a way to use my TP-K02 temp probe (K type rtd) with my DMM. Probe reads 15.2 ohms at 68 degrees F.
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jan 27, 2016 11:53 PM
This is from the manual
Probe Settings: RTD 2w, RTD 4w (default), Thermis2w, Thermis4w, TCouple (34465A/70A only)
I have confirmed that my 34461A do not have the TCouple probe setting in temperature messurement.
So the short answer is no.This may change in a future firmware upgrade (please don't hold your breath).
The longer answer is that you can use your probe, this table [http://www.pyromation.com/Downloads/Data/emfk_c.pdf] and messuring the DC voltage from your probe, lookup the temp in the table and you'r done. The K type probe messures 0V at 0C -V at negative temperature and +V on positive temperature (100C = 4.096mV and 37C = 1.489mV)
If you are stuck in the F scale of temperature you can use this table [http://www.pyromation.com/Downloads/Data/emfk_f.pdf]
More tables ? Try this [http://www.pyromation.com/Downloads/Data/temperature-tables.pdf]
Show 0 Likes
Retrieving data ...
2017Q1_SSO Simulation with Power Aware IBIS Model within ADS
Agilent 33522A - "Garfield.exe" error - must power cycle unit to clear
Application "garfield.exe" is the bane of my existence
33500B Series 3.05 firmware update error
IDC in ADS