to create and rate content, and to follow, bookmark, and share content with other members.
How to check sliding loads
Question asked by
on Aug 13, 2015
on Feb 11, 2016 by AOG
Show 0 Likes
Hi, apart from checking pin depth and mechanical functionality what would be the best way an user can check if a sliding load is working well. I want to check the loads of a 85052B.
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Feb 11, 2016 12:27 PM
When checking a sliding load you will begin with the basics like a visual check for any damage to the mating plane or elsewhere. After you set its pin depth to the applicable value, the next step would be to verify its return loss. This can be difficult if the load's return loss is greater than -46dB unless you have a VNA with corrected directivity much better than that. If so, you will need to measure the return loss (actually its resistive and reactive values from the Smith Chart) of the sliding load in each of its positions (usually 6 for HP/Agilent/Keysight sliding loads). From there you'll need to calculate the effective return loss of the sliding load. NIST has written several papers on how to do it. In the meantime I attached a spreadsheet you can use. Enter your six resistive and six reactive values on the RAW tab. There is a bunch of math going on with the next 5 tabs...you'll be interested with the final answer on the DATA tab. Hope it helps.
Sliding Load Calc 3.5 mm Scratch Sheet.xls
Show 0 Likes
Retrieving data ...
# Query for E4433B ESG-D Series signal generator regarding 2GFSK.OQPSK Signal generation
#Query regarding the E4406A Vector signal analyzer
Query regarding E4406A VSA Transmitter Tester, 7 MHz to 4 GHz
SiPro Differential S-Parameter Plots
Momentum in Virtuoso