to create and rate content, and to follow, bookmark, and share content with other members.
what is the minimum power that N5230A 240 can detect?
Question asked by
on Jun 24, 2007
on Jun 24, 2007 by Dr_joel
Show 0 Likes
If anyone knows, please tell me, I was reading those specification, but didn't find that. Thanks a lot!
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Jun 24, 2007 7:35 PM
You can find the appropriate entry under "test port input" in specification section of the help file. The help file is on the PNA, or you can find the online version here:
You can see it varies between about -112 dBm and -120 dBm, in a 10 Hz BW. Narrower bandwidth or increased average can improve the dection level as well. Option 245 (which can be added) will increase the level to -136 dBm best case in 10 Hz IF BW.
Please let us know your application, and we may be able to give you a better answer.
Show 0 Likes
Retrieving data ...
34461A DMM data logging with 34980A DAQ
Wide BandGap Semiconductors (WBG) - Why we should care.
How do I measure Chirpped Pulse's SSL and PCR
Equipment for Time-Resolved Chirp Measurements
Understanding Oscilloscope Trigger System Basics & Why You Should Care