to create and rate content, and to follow, bookmark, and share content with other members.
HP 8591C and HP 85721A
Question asked by
on Jun 15, 2013
on Jun 17, 2013 by MikeM
Show 0 Likes
Where I find the HP 85721A to download or buy?
No one else has this question
Mark as assumed answered
This content has been marked as final.
Show 1 comment
(Required, will not be published)
Aug 21, 2014 6:44 PM
All 8591C instrument shipped from HP/Agilent with the 85721A CATV measurement personality installed in them along with a memory card that had the software on it so that it could be reinstalled if necessary. Of course, over the years many of these have been lost.
Unfortunately HP/Agilent never provided a way for the user to download the software or load it into the instrument any other way that via a memory card, so I can't imagine you being able to restore it that way. We also don't have the ability to provide replacement cards any longer, so your only hope of getting one of these would be from a third party vendor or possibly someone here on the forum that has an extra. Of course, there are many different versions of this CATV personality, since there are many different TV formats around the world, and most use a different version of the 85721A (NTSC, PAL, SECAM, etc...) so be careful if you are going to purchase something...
Another possible solution may be to contact the Agilent service center near you to see if they might be able to load it for you if you were to send it into them, as they may still have a card for doing so when a servicing might require them to clear the memory of an instrument.
Show 0 Likes
Retrieving data ...
Budget functions in schematic
circuit component pin number does not match EM layout pin number
Simulation error:error detected by hpeesofsim in loading dynamic device code during netlist flattening.
ADS netlist – error caused by duplicate expressions in export
BenchVue Fails to Start